The Keysight A has a fixed output impedance hp a labview 50 ohms on the OUTPUT terminal and the A factory defaults set the display to show the output voltage when connected to a 50 Ohm load. The help file presents application programming examples, a hp a labview between instrument commands and driver functions, and detailed documentation of each function, with examples. Hi, I have installed the Instrument drivers for the HP A function generator fg , that were supplied by NI, and managed to get it working across serial. By default this is set to C:. REM” after the reset. The tread ended but nobody posted the solution. These VIs include the majority of the instrument driver template VIs, such as reset, self-test, revision, error query, and error message.
|Date Added:||18 September 2011|
|File Size:||40.72 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The HP A driver is located in a self-extracting archive file called hpa. It should be able to tell you what commands are really being sent.
HP A driver problems – Discussion Forums – National Instruments
HP A driver problems. The free IntuiLink Waveform Editor software will download waveform data from Keysight Infiniium, series, and series oscilloscopes and send the data to the a Keysight A, A, A, or A generator.
Enable Hp 33120a labview to view hp 33120a labview images. The alternative to using the NI Instrument Driver Finder is to go directly to the Instrument Driver Network and search for your instrument driver there. But I get an error message actually on the fg when the initialise vi is run in Labview. Most Active Software Boards: It includes a comprehensive on-line help file which complements the instrument manual.
Message 8 of Message 4 of Message 7 of The Initialize VI- the first instrument driver VI you call, establishes communication with the instrument. As a work hp 33120a labview, immediately after the init function, labviee the error query function to clear the instrument’s error buffer.
Hp a labview file can be found in 33120x vxipnpwin95hpa directory. 3320a
What is the proper termination character for the A? Can you download the values of the X and Y axis of a waveform obtained from the oscilloscope into a labvied hp 33120a labview to be use as the output?
Some older versions of the IB Libraries provided this hp 33120a labview file during installation. Remove the power cord. There are three ways to rack instruments of different depths side-by-side; for example, a Keysight A with a Keysight A. Message 5 of Make sure that for each successful call to the Initialize VI, you use a matching Close VI to avoid maintaining unnecessary memory resources.
Noticed that you are using serial port to communicate to instrument. Numerous Configuration VIs can exist, depending on the particular instrument.
If you do find a command that is causing the instrument to report an error, contact NI so that they can fix the bug. Message 1 of If you do not have ftp or web access, the driver can be obtained or by calling HP On-line Support hp 33120a labview The Keysight Hp 33120a labview has three forms of audible feedback. There are two possible termination characters that can be used for the A. The only hp a labview are that it does not have a soft front panel or a knowledge base file.
The reset might take hl out of SYST: The help hp 33120a labview presents application programming examples, a hp a labview between instrument commands and driver functions, and detailed documentation of each function, with examples.
Finally, the Simple Error Handler is called which indicates whether an error occurred. Most function generators have a built-in dc offset that can be added to any waveform.
How to Use an Instrument Driver in LabVIEW Tutorial – National Instruments
Would you please double-check the serial setting parameters, such as Baud Rate, Flow Controls and lavview. LabVIEW instrument drivers simplify instrument control and reduce test program development time by eliminating the need for you hp 33120a labview learn the complex, low-level programming commands for each instrument.
How do I delete a user-defined arbitrary waveform hp a labview memory? After the instrument has been configured, the Read VI is used hp 33120a labview read multiple points.